1. The emissivity of the DTR-1600 black body radiation source is as high as 0.998;
2. The opening is 50mm, which satisfies the calibration of most D: S thermometers.
3. The temperature control stability is not more than 0.3 ℃ / 10min. During the period when the blackbody radiation source is stabilized, the standard radiation thermometer monitors the target bullseye reading curve as a horizontal curve.
4. Good uniformity, effective temperature field up to 40mm, 0.2 ℃ @ φ30mm @ 1600 ℃, 0.6 ℃ @ φ40mm @ 1600 ℃.
5.The controller uses imported European brand 3504, 16 points correction, high temperature control accuracy.
6.The heating rate, the whole hundred points of heating plus the stabilization time is about 40min, to improve work efficiency.
7. Good repeatability, test is less than ± 0.25 ℃
8. The touch screen has a 30min heating curve record.
|Controller||Continental 3504 Thermostat|
|Ambient temperature||5～40℃/Constant temperature environment|
|Previous:DTY2005 intelligent pressure calibration device||next:DTLH-1 Low humidity intelligent temperature and humidity standard box|
Headquarter Address ：Tai'an High-tech Development Zone, Shandong Province
Telephon ：+86 0538-5089056 +86 13605384645 +86 13953856217
Fax ： +86 0538-5059718
After sale：+86 0538-5050959 +86 13605384645
Website ： www.amenaide.com
E-mail ： firstname.lastname@example.org email@example.com
Copyright? Tai'an Detu Automation Instrument Co., Ltd. Lu ICP 14024757-1